KEITHLEY 2000 LABVIEW DRIVERS FOR MAC
I don’t have the manual to confirm that though. You are right with the SRQ vi. So it would plot one reading per iteration loop. Still very new to LabView, so if you can help, assume I dont know anything about it. I have not yet seen any way to measure multiplie channels, do you know how that is done? Message 6 of Board index All times are UTC
|Date Added:||22 June 2015|
|File Size:||37.86 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. If so, do you have a serial or GPIB connection. This is a one shot keithley 2000 labview of mode. How do you have the keithley 2000 labview configured? Message 1 of The Configure Scan function will allow you to set the start and stop channels.
Keithley Series Native LabVIEW Driver Version | Tektronix
Is there a way I can read 2-wire resistance measurements from the as a function of time through labview in real time, 1 measurement keithley 2000 labview second? If you want a continous plot of data at this rate, Keihhley do not believe you will get there with this meter. So the plot would not keep up with this.
Plotting Keithley 2000 labview in real time from Keithley using Labview, Message 8 of Who is online Users browsing this forum: I then picked the driver for labview 8. Message 6 of Message 7 of The instrument will issue a service request when a reading is available. SRQ is a service request.
See how fast that is. You get readings at a rate of per second.
If you look at the Keithley Read single Example, if sets the instrument up for bus triggering and then the bus trigger command has keithley 2000 labview be issued. Labgiew sub vi that Keithley has provided called ke Data Read Single that is contained in a while loop will run only 5 times before hanging up. Message Edited by djconfused on I don’t have keithley 2000 labview manual to confirm that though.
After quite a bit of changes, I am still having the problem with the Read Single vi working for the first few times then pausing indefinitely I think. If you don’t want to issue a bus trigger, I believe you can set it up for immediate and then each keithley 2000 labview you call the read, it will return a keithley 2000 labview.
It can plot data in real time, but the rate is not so fast, labvoew fact, the rate is wrong: Message 9 of I have not selected a Keithley 2000 labview resource name, how is that done? I wrote RESGraph with a single reading in mind.
All posts 1 day 7 days 2 weeks 1 month 3 months 6 keithley 2000 labview 1 year Sort by: Additional NI Product Boards: Or let keithley 2000 labview know if you need it in another version of Labview. You need to select the correct resource. I noticed that if the Read Single vi was in a while loop and the Highlight Execution option was turned on, the Read Single would pause after three or four sucessful executions on first opening LabVIEW.
No registered users and 0 guests. Keitley are actually a few different drivers for the Keithley available at our driver site, http: I am using a GPIB connection to communicate with the multimeter.